But i need such code’s through that i m able to apply for my 5. I m facing increment problem only. And about all I can tell is that your sweep function is sort of working it does not start at the initial voltage and does not stop at the final voltage. How to add decimal point in out put data. Message 30 of
|Date Added:||13 September 2014|
|File Size:||52.83 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Message 26 of Most Active Software Boards: Why would you see if the function selected is greater than 4.
Here I m attaching my code In which Input voltage labbiew is there there we have to write command for input voltage i. Here i m having instrument but not able to solve the problem. And about all I can tell is that your sweep function is sort of working it does not start at the initial voltage and does not stop at the final voltage.
Tektronix and Keithley
Have you got code that updates the voltage range? Message 22 of Are you using the VIs which I found here, at ni. I need to work on this. But problem is each albview I’ve to mention.
Still now my code is correct only problem, commands how to give, its giving data for 1 volt but not the middle values, for that which command should I use, I tried all the necessary commands in my code. Message 23 of So may be it will difficult you to open.
Pabview 2 of So i want to finish my work with 5. So its becoming very tough for me to check where the error is occuring,when i run the program it will show ‘derr’means d error near to triggering.
Most Active Hardware Boards: Message 21 of Please show your latest VI and typical results for the critical range of Sir you just go through my attachment in that keithlet see V-Source I’ve mentioned Initial Voltage and Final Voltage middle increment window and if you have any correction then do it and send it to me sir. Posting a top level VI does not mean that you attach all of the subVIs.
Model Programmable Electrometer – Tektronix and Keithley
Message 10 of All the things are ok. This will pause the program. You should post the code you aer using now.
For hi-jacking thread, I was searching related to my problems, I found that one, so there only I asked my question. I know its difficult without labview 5. Message Edited by GerdW on Determine what needs to be done to the instrument in order to take the measurement. Labview VI’s for keithley meter. Message 8 of Message 3 of Nobody can help you on such vague information This code will generate the correct values that you would input to the instrument.